View and Download Escort A operation manual online. Count Multimeter. A Multimeter pdf manual download. Here you will find a large selection of new and used Escort A listed at special prices. Although some items match several categories, all of them are listed. Section 7. GPIB Remote Operation. Introduction. Description of the GPIB. GPIB Interface Parameters Setup. Commands Summary. 72 .
|Published (Last):||20 April 2013|
|PDF File Size:||20.45 Mb|
|ePub File Size:||8.47 Mb|
|Price:||Free* [*Free Regsitration Required]|
This manual contains information and warns which must be followed to ensure safe operation and retain the meter in safe condition. Refer servicing to qualified personnel.
The “trig” command may then be used to trigger a measure and get the related data. Releases the locking of the device’s keyboard. The power-up configuration set at the factory is shown in Table The correct fuse ratings: Vac and frequency of the input signal you are measuring. To display DC voltage in primary display, and AC voltage in secondary display or vise versus while testing a rectifier circuit, user may check the DC voltage supplied and its AC ripples by taking a single meter.
Do not apply a voltage exceeding V peak between and COM input terminals. It is presently not possible to handle more than 1 mmA multimeter per computer. Rear Panel The rear panel shown in Figure contains a line fuse, the power-line cord connector, an RS interface connector, and a cutout for IEEE interface optional connector.
Operating in Explosive Atmospheres The meter does not provide explosion protection for explosive gasses or arcing components. DBM – dBm calculation. Bit set to 1 when response to query placed in output buffer. To avoid personal injury, do not remove the cover or panel. Press to enter Refresh Hold mode.
Escort 3136A Manuals
The communication parameters are set to bauds, no parity, 8 databits, 1 stop bit. Sets the brightness of the displays. Overload Alert The meter has provided an overload alert for voltage and current measurements.
OPC – Operation complete. R2 command After executing R2 command, the meter will return the existing readings of secondary display. Page 12 Rear Panel The rear panel shown in Figure contains a line fuse, the power-line cord connector, an RS interface connector, and a cutout for IEEE interface optional connector.
CURRent Set the current measurement. It also explains the detail information of all IEEE If RS of the meter is under print-only mode, the meter will print out the measured data when the measurement cycle is completed.
Selecting Trigger Mode once over the maximum display of IEEE interface Optional provides full programmability. Page 41 … … Press DOWN button again tothen the setting value will be subtracted automatically. Set the specified display in Vdc Direct Current Voltage mode, within the specified voltage range. The Auto-ranging Relative mode or auto-ranging dynamic record will be closed esxort percentage function is set.
Moreover, a TCL session is created for handling the communication and the related data.
Escort A Manuals
Page 38 Press to enter Refresh Hold mode. Turns “off” the trigger mode. To display AC current in primary display and DC current in secondary display or vise versus while testing a rectifier circuit, user may check the DC current component and its AC ripples by taking a single meter. Selects the source of the start trigger signal. Opens the serial port that number from 1 to is specified.